KB9BVN - <kb9bvn@...>
toggle quoted messageShow quoted text
I think I found it....google search for Qrpometer Docs came up with this http://www.4sqrp.com/kits/QRPometer/qrpometermanualv4.pdf
From 2012....C3 is listed as .001 and marked as 102.
He may be using an old doc? Or maybe he has an old kit?
On Sun, Oct 20, 2019 at 8:04 PM nm0s_qrp via Groups.Io <firstname.lastname@example.org> wrote:
Where are you getting that C3 has a value of 0.001 uF? None of the documentation has this value. In the instruction sheet parts list, assembly instructions, and schematic, the value of 0.1 uF is specified. Please follow the instructions, and use the correct value.
73 de KB9BVN