Re: Qrpometer calibration
toggle quoted messageShow quoted text
The Qrp o meters have a history of needing a lot of heat on the load resistors.
Try cranking up the heat to high, then reflow them and let em soak.
It’s worth a try.
On Thu, Feb 25, 2021 at 8:43 PM AG4CC <telegraph@...> wrote:
Check out the 4SQRP website at 4sqrp.com